1. Working power supply: AC220V, 50Hz;
2. Horizontal rotation angle of terminal box body: 0 ~ 340 °
(3) Net load weight ≤ 80 kg;
(4) Instrument platform: 550 mm-400 mm in the second layer (adjustable height), with round-angle anti-collision design;
(5) Gas interface configuration (national standard dioxygen absorption or configuration according to the requirements of the hospital): a. The interface is different in color and shape and has the function of preventing connection errors; b. The number of insertions and extractions is more than 20,000 times;
7. Power sockets: 8, 220 V and 10 A; one for each telephone and network interface
(6) Equipotential grounding terminal: one;
7. One stainless steel infusion pole;
(8) The main material shall be high-strength aluminum alloy profiles.
(9) Electrostatic spraying is used for surface treatment;
10. Suction top installation, stable and firm.